Advanced Chemical Sensors, Inc.
Technical Information Sheet
Halogenated Anesthetics Vapor Monitor
Catalog No. H-10-3
Simultaneous measurement of waste anesthetic gases: Sevoflurane, Isoflurane, Enflurane, Halothane,
ACS Vapor Monitor can be worn as a badge near the breathing zone
to measure personal exposure, or placed in a room to measure area
Occupational safety standards are based on personal exposure
monitoring of each employee who may be exposed to the vapor as a
regular part of their job. Periodic monitoring of personnel and room
concentrations is recommended
||Activated Carbon Molecular Sieve Adsorbent
|Method of Analysis:
||OSHA Method 29
|Recommended Sampling Time:
||30 minutes to 8 hours
|Minimum Level of Quantitation:
||0.02 ppm for 15 minutes (using Lab Blank)
0.01 ppm for 15 minutes (using Field Blank)
||Compounds that may adsorb on the collection medium will not will not interfere with the analysis.
|Effect of Temperature:
||Less than 10% for each 10 °C variation from 24 °C
|Effect of Humidity:
||No effects detected at 30% RH to 92% RH.
|Overall System Accuracy:
Better than +/1 15% at 2 ppm for 8 hours
||Maximum shelf life one year.
|Recommended Holding Time:
||Two weeks prior to analysis.