Advanced Chemical Sensors, Inc.
Technical Information Sheet
Chloroform Vapor Monitor
ACS Vapor Monitor can be worn as a badge near the breathing zone
to measure personal exposure, or placed in a room to measure area
Occupational safety standards are based on personal exposure
monitoring of each employee who may be exposed to the vapor as a
regular part of their job. Periodic monitoring of personnel and room
concentrations is recommended
||Activated High-Purity Carbon Molecular Sieve
|Method of Analysis:
||US OSHA Method 5
|Recommended Sampling Time:
||15 minutes to 8 hours
|Minimum Level of Quantitation:
||0.2 ppm for 8 hours
7 ppm for 15 minutes
||No interferences are known.
|Effect of Temperature:
||Less than 10% for each 10 °C variation from 24 °C
|Effect of Humidity:
||No effects detected at 30% RH to 92% RH.
|Overall System Accuracy:
||11% at 10 ppm for 8 hours
16% as 50 ppm for 15 minutes
||Maximum shelf life one year.
|Recommended Holding Time:
||Two weeks prior to analysis.