Advanced Chemical Sensors, Inc.
Technical Information Sheet
Glutaraldehyde Vapor Monitor
Catalog No. G-10
ACS Vapor Monitor can be worn as a badge near the breathing zone
to measure personal exposure, or placed in a room to measure area
Occupational safety standards are based on personal exposure
monitoring of each employee who may be exposed to the vapor as a
regular part of their job. Periodic monitoring of personnel and room
concentrations is recommended
||2,4-DNPH Coated High-Purity Silica Gel
|Method of Analysis:
||High Pressure Liquid Chromatography
||OSHA Method 64
|Recommended Sampling Time:
||15 minutes to 24 hours
|Minimum Level of Quantitation:
||0.02 ppm for 15 minutes (using Lab Blank)
0.01 ppm for 15 minutes (using Field Blank)
||Other aldehydes such as Formaldehyde will not interfere.
No interferences are known.
|Effect of Temperature:
||Less than 10% for each 10 °C variation from 24 °C
|Effect of Humidity:
||No effects detected at 30% RH to 92% RH.
|Overall System Accuracy:
11% at 0.2 ppm for 15 minutes
||Maximum shelf life 6 months.
|Recommended Holding Time:
||Two weeks prior to analysis.