Advanced Chemical Sensors, Inc.
Technical Information Sheet
Nitrous Oxide Vapor Monitor
Catalog No. N-10
ACS Vapor Monitor can be worn as a badge near the breathing zone
to measure personal exposure, or placed in a room to measure area
Occupational safety standards are based on personal exposure
monitoring of each employee who may be exposed to the vapor as a
regular part of their job. Periodic monitoring of personnel and room
concentrations is recommended
||Activated Sodium Alumino Silicate Molecular Sieve Adsorbent
||NIOSH Method 6600
|Recommended Sampling Time:
||30 minutes to 8 hours
|Minimum Level of Quantitation:
||1 ppm for 1 hour
||No interferences are known.
|Effect of Temperature:
||Less than 15% between 56 °F and 111 °F
|Effect of Humidity:
||No effects detected at 30% RH to 92% RH.
|Overall System Accuracy:
Better than +/-16% at 25 ppm for 8 hours
Coefficient of Variation = 0.039
Found to be the only accurate monitoring badge by
the American Dental Association.
||Maximum shelf life one year.
|Recommended Holding Time:
||Two weeks prior to analysis.